 Home /
 Resistance /
 Decade Resistor /
 HATSLR Low Resistance Transfer Standard
HATSLR Low Resistance Transfer Standard
Double click on above image to view full picture
Photo represents example of series. See datasheet for more details on any specific model.
 Values from 1 Ω  100 kΩ
 Initial Accuracy... As good as 10 ppm of nominal value
 Transfer Accuracy... ± (1 ppm + 0.1uΩ at parallel value) for 10:1 & 100:1 ratios
 Long Term Accuracy... ± 10 ppm/year
 Calibration Accuracy... ± 5 ppm
Estimated Delivery is 5 to 8 weeks
Repair/Calibration Quote Technical Support RequestThe HATSLR resistance transfer standards consist of 12 equal resistance steps. These steps are available in increments of 1 Ω, 10 Ω, 100 Ω, 1 kΩ, 10 kΩ, and 100 kΩ. The HATSLR provides fourterminal connections for each step or for series and parallel combinations of these resistance steps using the optional SeriesParallel (SPC102), Parallel Compensation Networks (PC101) and shorting bars (SB103).
Description
The HATSLR meets or exceeds all of the requirements for resistance transfer standards in precision measurement applications. It is easily configured to transfer resistances up or down a decade from their initial resistance value. When used with the connecting networks and shorting bars, it provides 1 ppm transfer accuracies.
Each transfer standard contains twelve equal value precision resistors connected in series by specially designed true 4terminal junctions. These special junctions assure that a 4terminal measurement of a series of resistors agrees with the sums of the individual resistors in the series. Accurate parallel connections can be made with the Parallel Compensation Network and the Shorting Bars connected to the junctions.
These standards can be connected to provide three decade values: 10 resistors in series, 10R: 9 resistors in series — parallel, 1R: and 10 resistors in parallel, R/10. The part per million accuracy is assured as the series value is equal to 100 times the parallel value to better than 1 ppm. The series — parallel value relative to either the series value or the parallel value can be found to better than 1 ppm by making a 1:1 comparison with the remaining tenth resistor and a simple calculation.
The accuracy and precision of the individual resistors also make the HATSLT transfer standards ideal for use as a multivalue standard resistor or reference voltage divider.
Advantages of Transfer Standards
In order to perform calibrations with a high degree of accuracy, reference standards must be employed at every range or decade of the measuring or calibration instrumentation. Clearly, this can be difficult and costly since these standards must be highly stable and their precise values must be known with a high degree of certainty and sufficient resolution.
To minimize the cost and difficulty, more practical means of performing such calibrations is to use transfer standards.
If one has a single standard that is calibrated by a national laboratory, one can then compare the transfer standards to the certified standard by ratio techniques. See our Technical Applications section for a full tutorial.
Resistor Type: Wirewound, hermetically sealed, low inductance.
Step Size: 1 Ω, 10 Ω, 100 Ω, 1 kΩ, 10 kΩ, or 100 kΩ.
Initial Accuracy: <±15 ppm for 1 Ω, 10 Ω steps; <±10 ppm for 100 Ω through 100 kΩ steps.
Long Term Accuracy: <±10 ppm/year;
Transfer Accuracy: ±(1 ppm + 0.1 μΩ) for 10:1 and 100:1 ratios for 1 Ω, 100 Ω, 1 kΩ, 10 kΩ, and 100 kΩ steps;
±1 ppm for 10:1 and 100:1 ratios for 10 Ω step.
(Transfer accuracies apply when SB103, fixtures are used)
Matching Accuracy: within 10 ppm for 1 Ω, 10 Ω steps; within 5 ppm for 100 Ω through 100 kΩ steps.
Matching Temperature Coefficient: within 5 ppm/°C for 1 Ω step; within 1 ppm/°C for ≥ 100 Ω steps and for HATSLRTC10 within 2 ppm/°C for HATSLR10 only
Matching Calibration Accuracy: <10 ppm for 1 Ω, <5 ppm for all others
Temperature Coefficient: ±1 ppm/°C for HATSLRTC10 (low TC version), ±10 ppm/°C for HATSLR1;, ±3 ppm/°C for HATSLR10;, ±2 ppm/°C for HATSLR100 through 100K.

NIST Traceable Calibration Cert (Included)
+ $0.00

ISO17025 Accredited Calibration with Test Data
+ $150.00

PC101 Parallel Compensation Network
+ $1695.00

Shorting Bars
+ $1295.00

SPC 102 Series Parallel Compensation Network
+ $1695.00

RM100 General Purpose Test Equipment Rack Mount Shelves
+ $275.00

TLK100 Deluxe Set of Testing Leads (17 pcs)
+ $94.00
Product Categories

http://www.ietlabs.com/decaderes.htmlResistance

http://www.ietlabs.com/decaderes/decaderesistor.htmlDecade Resistor

http://www.ietlabs.com/decaderes/resistancestandard.htmlResistance Standard

http://www.ietlabs.com/decaderes/resistancemeter.htmlOhmmeter


http://www.ietlabs.com/capacitance.htmlCapacitance

http://www.ietlabs.com/capacitance/decadecapacitor.htmlDecade Capacitor

http://www.ietlabs.com/capacitance/capacitancestandard.htmlCapacitance Standard

http://www.ietlabs.com/capacitance/capacitancemeterbridge.htmlCapacitance Bridges


http://www.ietlabs.com/inductance.htmlInductance

http://www.ietlabs.com/inductance/decadeinductor.htmlDecade Inductor

http://www.ietlabs.com/inductance/inductancestandard.htmlInductance Standard

http://www.ietlabs.com/inductance/inductancemeter.htmlInductance Meter


http://www.ietlabs.com/decadebox.htmlDecade Boxes

http://www.ietlabs.com/rtds.htmlRTD Simulators

http://www.ietlabs.com/microohmmeter.htmlMicroOhmmeters

http://www.ietlabs.com/ohmmeters.htmlMegohmmeters

http://www.ietlabs.com/digibridges.htmlImpedance Bridges and Meters

http://www.ietlabs.com/lcrmeter.htmlLCR Meter

http://www.ietlabs.com/variac.htmlVariacs

http://www.ietlabs.com/stroboscope.htmlStroboscopes

http://www.ietlabs.com/soundlevelmeter.htmlAudio

http://www.ietlabs.com/voltagedivider.htmlVoltage Dividers

http://www.ietlabs.com/decadevi.htmlVoltage & Current Sources

http://www.ietlabs.com/gr874connectors.htmlGR 874 / Test Leads

http://www.ietlabs.com/dielectriccells.htmlDielectric Test Cells

http://www.ietlabs.com/scrollingcatalog.htmlScrolling Catalog
 Home
 Products
 Calibration & Repair
 Company
 Contact Us
 Terms/Privacy Policy
 Request Quote
 Technical Support
 Site Map
PHONE: (516) 3345959 FAX: (516) 3345988 SALES@IETLABS.COM